CORRELATION ANALYSIS OF
ACCUSENSING MAS AND ICP-OES
FAST AND ACCURATE –
A total of 35 groundwater samples were detected for arsenic in field by MAS. Afterwards the water samples were sealed to laboratory for ICP-OES analysis. Correlation analysis between MAS and ICP-OES figures that the correlation coefficient(r) equals 0.9818.
PROVEN IN FIELD
USER-FRIENDLY, FAST AND ACCURATE
|Conc. range||21.6~399.0 µg/L|
|Product / Model||MAS (Metals Analysis System) / G1|
|Main Function||Detecting metal ions in liquid samples|
|Principle||Anodic Stripping Voltammetry, ASV|
|Features||1. Calibration-free and pretreatment-free with using
ACCUSENSING MULIETM sensors.
2. Digital read-out display simultaneously with
electrochemical spectrum within a few minutes.
3. Manual correction for peak identification if need.
4. 100 data storage for each metal.
5. Metal mode is switched by changing different smart chips.
|Operation Panel||5 inch touch panel|
|Size / Weight||size : 187(L) x 107(W) x 43(H)mm
weight: 0.65 kg(device) / 2.8 kg(Including a case and accessories)
|Power Supply / Battery||5V 1600 mA / 3700 mA Li-Battery|
*Precision and linearity are tested by ACCUSENSING As100 sensors